For high noise emissions, especially for large cable lengths, the use of shielded cables is recommended. AB 1-point calibration Temperature: As defined in the Terms and Conditions of Sale, Pall Europe does not accept any liability whatsoever for the remote collection, shipping costs or packaging of returned goods. Sets the switching hysteresis that should be maintained each time the signal falls below the upper switching limit. If this equipment does cause harmful interference to radio or television reception, which can be determined by turning the equipment off and on, the user is encouraged to try to correct the interference by one or more of the following measures – Reorient or relocate the receiving antenna. Simple Installation The Water Sensor enclosure is designed in modular form for safe and convenient installation. Value of the saline solution or display of HUMOR 20 4 By clicking on “Save”, the humidity reading of the transmitter will be adjusted to the reference humidity.
Bottom and centerpeice of the housing shall be grounded! Deletes groups within a tree. Socket for optional display: Follow the instructions of the dialogue menus to set the desired language and all other parameters for installation. For this reason, the manual must be read before start-up. By not following these instructions injuries of persons or material damage could happen. This equipment generates, uses and can radiate radio frequency energy and, if not installed and used in accordance with the installation manual, may cause harmful interference to radio communications.
Fast and easy calibration for accurate measurement results at a defined working point humidity point. The described products can be improved and changed at any time without prior notice. This equipment generates, uses and can radiate radio frequency energy and, if not installed and used in accordance with the installation manual, may cause harmful interference to radio communications.
The default factory output setting is for current. Wz08 1-point calibration Humidity: Deletes groups within a tree. The measured values will now appear in both input fields.
Ball valve set C08 option: Check that the probe is not free to slide in the gland — tighten further if it can be moved by hand. Because of developments in technology these data or procedures may be subject to change.
Ball valve set B08 option: Provides information on the date of the last humidity calibration, but only if the configuration software was used.
Customers are responsible for the costs of shipping goods to the Central Collection Point and Pall Europe Ltd accepts no liability whatsoever for any such costs.
The document may contain technical inaccuracies and typographical errors. At the end of the routine, the software is installed and the Readme file or the program will be automatically opened. The limits must fall within the operating range indicated above. Pall and are trademarks of Pall Corporation. Uninstall the WS08 Configurator by clicking on the appropriate button and then reinstall or upgrade.
Opens a file for a new tree. All costs associated with the return of WEEE will be born by the customer. Each transmitter is assigned a unique network address at the factory for precise identification.
Contact Pall or your local supplier. In a separate window, the measured values will appear in both input fields 3 Replace the value in the input field “Humidity Reading” with the reference humidity.
Before any reinstallment or upgrade the older version must first be uninstalled the User will be notified during the installation routine and the process will be interrupted automatically. Calibrations performed directly on the circuit board are not recorded!
NEW: Pall WS08 Series Water Sensor Data Sheet Features
Where facilities exist, component parts of the unit may be recycled. Following functions are available: Displays the preferences for all transmitters that have been set-up.
Two relays are provided which can be configured using the configuration software supplied Note: Tighten sufficiently to seal the joint. Precautions therefore need to be taken when installing or removing the probe to ensure that it is not ejected from the system at high velocities. Note that during calibration operations using the buttons on the circuit board, the significance of the LEDs being on, or flashing, is different from the above. Relays are not available with RS option.
Add this document to saved.
Datasheet: Pall Ws08 Series Water Sensor
Note that a force of at least 1 kgf per bar of pressure 1. Reads and displays the configuration parameters of the selected transmitter. Provides information on the date of the last temperature calibration, but only if the configuration software was used.
Tightening the gland-nut should eliminate the small leak which may occur during installation. Firmly and steadily press the probe through the gland until the tip seres in the flow of fluid.
WS08 instruction booklet Pall WS08 Series Water Sensor
Slide the probe through the gland to locate the probe tip in the fluid flow, and tighten the gland lock-nut onto the nipple to a torque of 30Nm 22 lbf ft. The water sensor is not to be used in water based fluids or aerospace phosphate ester hydraulic fluids.
This is essential if the operating pressure is above about 4 bar 58 psiand recommended for lower pressures. The preferences may also be changed here. The document may contain technical inaccuracies and typographical errors.
Primary Fluid Power – New WS09 Portable hand-held Water Sensor
Socket for optional display: Selects the interface for connecting the transmitter to the network. Select the desired measured variable. The WS08 Water Sensor can display the highest and lowest measured value recorded since the last reset. These changes will be implemented in later versions.
In addition, the manual is for all personnel who require knowledge concerning transport, setup, operation, maintenance and repair. Part numbers appearing in this manual are protected by the Wws08 of Pall Europe Limited. The “Factory Calibration” button is located on the Calibration Tab see photo at the start of section 5.